We are often asked whether shakers that are continuously operated in climate chambers are influencing the temperature.
Heidolph has tested several shakers in Memmert climate chambers under real conditions.
These tests demonstrate: Even if several powerful shakers are operated synchronously inside a climate chamber, the temperature does not rise or rises only slightly. A Memmert climate chamber can be combined with Heidolph shakers to make up a powerful incubator system. The temperature deviations measured in the medium are within the tolerated margin.
Devices under test:
Representative of the Heidolph shaker portfolio, two devices of different dimensions were used in the test (only marginal deviations to other Heidolph shakers): Unimax 1010 and Promax 2020
The temperature was measured in the medium.
|1l-Erlenmeyer flask||800 ml water|
|Shaking frequency||80 rpm|
|Target temperature heating chamber||37 °C|