Yes, the rotary evaporator can be flooded with inertgas via the ventilation valve of the vacuum box.
In addition, a external pressure reducer is required to not exceed 0.2 bar overpressure (overpressure shutdown of Hei-VAP Precision).
Heidolph Korea Ltd.
#1309, 5, Digital-ro 26-gil, Guro-gu, Seoul, Republic of Korea
m: +82 10 91748838
© 2021 |
Heidolph Instruments GmbH & CO. KG
| Terms and Conditions
| Export Declaration